If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
Difference between revisions of "2440"
|Line 20:||Line 20:|
*2 ns Glitch Capture
*2 ns Glitch Capture
Revision as of 00:46, 15 December 2018
The Tektronix 2440 is a two-channel 300 MHz digital sampling oscilloscope from the late 1980s. The maximum sample rate is 500 megasamples per second, per channel.
- 500 MS/s Digitizing
- 300MHz Bandwidth
- 2 ns Glitch Capture
- Autoprobe Function With P6137 Probe
- 300MHz X-Y Bandwidth At Probe Tip
- Automatic Measurements
- Extensive Triggering Capabilities
- Printer/Plotter Output
- On-screen HELP Text Describing Every Front-Panel Control
Operation of the 2440 (and 2432 and 2430): A/D-converter and CCD hybrids may suffer irreparable damage within a few minutes if operated out of the metal enclosure without extra cooling.
|Bandwidth||300 MHz, Limit: 20, 100, Full|
|Deflection||Two Channels With Simultaneous Acquisition with sampling rate: 500MSa/s Each. CH 1&2: 2mV/div to 5V/div in 1-2-5 sequence|
|Auto Scale Factor||On-Screen Readout indicates probe tip deflection factors for 1X, 10X, 100X and 1000X probes|
|Input Impedance||1M, 15pF|
|Sweep Speeds||Modes A, A Intensified and B :2ns/div to 5s/div in 1-2-5 sequence|
|Auto Measure||Simultaneously measures up to four parameters on one or more displayed waveforms, updated a 3hz intervals or display a snapshot of 20 parameters of waveform target|
|Acquisition Modes||Normal, Envelope (glitch capture), Average|
|Display Modes||CH1, CH2, invert, add, multiply, X-Y|
|Cursors||Volts, Time, Volts At Time, 1/Time, Slope|
|Functions||A Mode - Auto Level, Auto/Roll, Normal and Single Seq, B Mode - Trigger After Delay, Runs After Delay and many more too long to list here|
|Probes||Two Passive Voltage 10X 1.3m P6137(400 MHz) Probes|
|Power||160 W, 115vac: 90 to 132vac, 230vac: 80 to 250vac, 48 to 440hz|
- 03 - Word Recognizer (P6407)
- 05 - Video Waveform Measurement System
- 11 - Two probe-power connectors on the rear panel