If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
The S-1 sampling plug-in head provides a single GR-874 50 Ω input with a 350 ps rise-time.
The S-1 can be used with the 7000 series of sampling plug-ins, as well as the 3S series of sampling units.
|Rise time||350 ps|
|Input amplitude||−1 V to +1 V with ≤1 Vp-p signals|
|Maximum input||±5 VDC or sine of 10 Vp-p up to 10 MHz|
|Input impedance||50 Ω (GR-874)|
|Noise||< 2 mV|
|Features||trigger pickoff for internal triggering|
The S-1 uses a two-diode sampling gate which has relatively high "kick-out" (leakage of the strobe signal out of the input).
The manual cautions,
Operating the sampling head without the input connector terminated by a 50 Ω resistor or coaxial cable will cause a vertical shift of the zero signal baseline by a few millivolts. This is caused by the strobe kickout signal being reflected from the open input connector, and orriving bock at the sampling bridge while the bridge is conducting.
7S11 with S-1