If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
The Tektronix Type 1A5 is a differential amplifier plug-in for 500-series scopes.
|Bandwidth||50 MHz down to 5 mV/div, 40 MHz at 1 mV/div|
|Deflection||1 mV/div to 20 V/div, 1−2−5|
|Common-mode rejection||20,000:1 at low frequencies, deteriorating to 200:1 at 20 MHz|
Like most Tektronix differential plug-ins, the 1A5 has three modes: VA−VC, VA−VB, and VC−VB.
VC is controlled by a ten-turn potentiometer and can vary from −5 V to +5 V (0.5 V on the 5 V/div to 20 V/div ranges).