2465 Series PSU Repairs
The power supply in the 2465 series scopes is a switch mode unit.
For the A and B variants, it is known that the markings for C1115 and C1132 are mixed up in all Tektronix documentation. Be wary of this when replacing these components. For the original 2465 (not A or B), the service manual is correct.