It is a complete sampling system unlike, for example, the 7S11, which requires a separate sampling timebase plug-in to provide triggering and sampling pulse generation.
|Rise time||350 ps (1 GHz bandwidth)|
|Trigger bandwidth||100 MHz (Norm/Auto trigger), 1 GHz (HF sync)|
|Vertical deflection||2 mV/Div to 0.5 V/Div in 1—2—5 sequence|
|Sweep rate||100 µs/Div to 100 ps/Div in 1—2—5 sequence|
|Input impedance||50 Ω|
|Maximum input voltage||5 V peak|
The 5S14N contains two samplers, trigger and sweep circuitry, and circuitry to interface it with the 5000-series mainframe in which it operates. The mainframe provides the 5S14N with power. As a combined horizontal/vertical plug-in the 5S14N sends the mainframe both horizontal and vertical signals.
There are two 1.35 V mercury button cells, BT1 and BT2, in the sampler circuit. They act as floating bias sources, so if a 5S14 stops working it may be not defective apart from the dead batteries.
The sampler used by the 5S14 is a two-diode design. Each of the two input channels has its own sampler.