If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
The Tektronix 7250 is a digitizing oscilloscope with 6 GHz bandwidth, optimized for digitizing single-shot events. It uses a scan converter CRT for digitizing. From the scan converter, the trace is transferred to memory boards containing conventional RAM.
The captured waveform can be transferred to a computer via a GPIB interface or can be viewed on the built-in monitor, which is a conventional computer-style CRT display and has menus.
At 60 kg (132 pounds), it is the heaviest one-piece oscilloscope ever sold by Tektronix. It was made by Intertechnique in France where it was sold as the IN7000, and rebranded by Tektronix for sale in the United States.
|— 7250 mainframe —|
|Bandwidth||6 GHz (measured as -3 dB @ 7 GHz, -6 dB @ 9.6 GHz, and -8.5 dB @ 14 GHz in Haas, Warman, van Ewijk, Rev. Sci. Instrum. 60 (3), March 1989)|
|Rise time||50 ps|
|Resolution||Vertical, 11 bits. The digitizer captures 512 samples in one shot.|
|Sweep||1 μs/Div to 50 ps/Div (1-2-5), equivalent to max. ~1 terasample per second|
|Sensitivity||fixed 500 mV/Div (no vertical amplifiers in the signal path)|
|Max. input||2 kV spikes < 1 μs|
|Triggering||External only, requires 50 ns pretrigger signal|
|— 7250 Opt. 01 External Delay Line —|
|Rise time||≤75 ps|
|Input / output||50 Ω N connectors|
|Trigger pickoff||20 dB attenuation, 300 ps rise time, BNC connector|
- Tektronix 7250 advertising video, 1987 (VintageTEK Museum @ YouTube)
|Opt. 01 Ext. Delay Line||Catalog price||$12,500||$12,500|
The 7250 has two 3.6 V ½AA lithium batteries on each memory board. With time, these batteries die, which will cause the 7250 to fail its power-on self test. New replacements are still (2011) widely available.