For operation, a 7T11 or 7S12 plug-in is needed, to which the 7S11 connects through contact strips on the right side of the plug-in. The control signals are routed to a similar strip on the left of the module for another 7S11 in a dual-channel configuration. An alternative configuration consists of a pair of 7S11s, one in the right vertical and the other in the A or only horizontal bay, for an X-Y display in free-running mode.
|Deflection factor||2 to 200 in 1-2-5 sequence, unit as labelled on the sampling head (usually mV, except for the optical-input S-42 scaled in mW)|
|DC Offset range||+/- 1 V|
|Offset output||10 × offset voltage, source impedance 10 kΩ|
|Vertical signal output||200 mV/Div, max. 2.4 Vp-p, source impedance 10 kΩ|
|Delay range||at least 10 ns|
|Weight||0.9 kg (3.25 lbs)|
The 7S11 does not contain integrated circuits (except dual transistors).