If you get the "Sorry, the file cannot be displayed. There seems to be a technical issue ..." error when trying to view an image, a workaround is to click on the cog icon in the upper right side of the media viewer, and then select "Disable Media Viewer". We are aware of the problem and are looking for a solution.
The Tektronix P6106 is a passive 250 MHz ×10 probe for 1 MΩ inputs. It was available with 1, 2 or 3 meters of probe cable. The 6106 has a coding pin for scope readout ×10. Color coding on socket tells, how long is their cable. Orange means 2m, Blue means 1m.
|Attenuation||10 × ±3% with scope input at 1 MΩ ±2%|
|Compensation range||15 pF to 24 pF|
|Bandwidth||>250 MHz (>150 MHz for 3 meter cable). Using the 1 meter probe with a scope having at least 325 MHz bandwidth results in a system bandwidth of 300 MHz typ.|
|Input capacitance||10.5 pF (1 m), 13 pF (2 m), 15.5 pF (3 m)|