Difference between revisions of "P6056"
m (Text replacement - "Category:50 Ohm Oscilloscope Probes" to "Category:50 Ω Oscilloscope probes")
|Line 17:||Line 17:|
* [://w140.com/tek_P6056_P6057.pdf Tektronix P6056 and P6057 Manual (PDF)]
Revision as of 14:07, 23 January 2021
The Tektronix P6056 is a 3.5 GHz ×10 probe for 50 Ω-input scopes, released along with the 7A19 amplifier in 1972.
It has an Identify switch and readout contact.
Its sister model P6057 has ×100 attenuation.
|Bandwidth||DC to 3.5 GHz|
|Rise time||100 ps|
|Input impedance||500 Ω // approx. 1 pF|
|Scope Input||50 Ω|
|Features||Readout Pin, ID Button|
This high performance device has a fragile resistor assembly in the
probe head, subject to failure in rough usage. Use with slip-on tip
whenever possible. One replacement part is provided with the package.
Contact nearest Field Office for additional parts (P/N 307-0330-00), if
required. For replacement and Calibration Procedure, refer to Pg. 3 & 4