The task of designing the R293 was inherited and completed by Murlan Kaufman.
The R293 contains an analog programmable power supply and an analog programmable pulse generator. The pulse is produced by an avalanche transistor fed through a pulse shaper made of diode clamps and snap-off diodes.
The output pulse amplitude is variable from 6 V to 12 V. Overshoot of the output pulse is less that 3%, rise and fall time are 1 ns or less.
The service procedure in the manual specifies the use of a 661 for adjustment of the pulse shaper.
Some instances of the R293 have an RF connector on the front panel while others have all of their connector on the rear panel. MOD 703M puts all input and output connectors of the R293 on the rear panel, for use in automated testing systems.
An R293 is normally programmed from an R250.
The power supply output connector mates with a Bendix PC06A-8-4P-SR connector.