The Tektronix TG501 is a time mark generator plug-in for the TM500 system.
Note the output must be terminated into 50 Ω or there may be no signal at all.
There is also a TG501A.
- Markers: 5 s to 1 ns in a 1,2,5 sequence
- Marker Amplitude (into 50 Ω): ≥1 V peak on 5 s through 10 ns; ≥750 mVp-p on 5 ns and 2 ns; ≥200 mVp-p on 1 ns (separate output)
- Trigger Output Signal: Slaved to marker output from 5 s through 100 ns. Remains at 100 ns for all faster markers
Internal Reference Standard (1 MHz) Option 1 (5 MHz) Stability 1E-5 5E-7 Long-Term Drift 1E-5 per month 1E-7 per month Setability 1E-7 5E-9
- External Reference Input: 1 MHz; 5 MHz; 10 MHz; TTL compatible (internally hard wired so output frequencies counted down to 1 MHz)
- Error Readout Range: to ±7.5%
- Error Readout accuracy: One digit (±0.1%-points)
- Net Weight: 1 kg
- Dimensions: 66.8 mm × 285.3 mm × 125.9 mm (w/d/h)
- Check for +15 V and adjust if not between 14.3 V and 15.7 V
- If +5 V is outside 4.94 V − 5.46 V, check U626 (LM741) − old opamps are prone to failure.
- A lot of ICs are placed in non gold plated IC sockets. Contact problems can occur. Removing and inserting ICs helps.
- A1 board should be mounted to frame only with washers inbetween, especially near the rear connector. Shorts to conductor tracks can occur otherwise.