The Tektronix 7A29 is a 1 GHz single-channel vertical plug-in for 7000-series scopes. It is primarily intended for the 7103 and 7104. Like with the 7A19, a variable signal delay was available as an option. The 7A29 was designed by John Addis.
The 7F10 uses the 7A29's amplifier board after the optical/electrical converter.
|Bandwidth||1 GHz (in 7104)|
|Input impedance||50±1 Ω, VSWR <1.2:1 from DC to 1 GHz (10 mV/Div: 1.4:1, Opt.4: 1.5:1)|
|Max. input||max. 10 VRMS or 1 Ws pulses not exceeding 50 V; max. 100 V additional DC component in AC mode|
|Deflection||10 mV/div to 1 V/div, 1-2-5, ±2%; variable 2.5:1|
|Recommended probes||P6056 (×10), P6057 (×100)|
|Options||Option 4: variable delay (1 ns)|
It uses a "feed-beside" scheme implemented with standard low-frequency operational amplifiers to compensate for low-frequency anomalies in the frequency response that are attributed to transistor heating.
The input connector is a special Tek-made part that is exactly 50 Ω. Typical BNC connectors are around 52 Ω.
The 7A29 does not use the ±50 V supplies provided by the mainframe.
Although the 7A29 has a VARIABLE gain control, this control has the undesirable side-effect of also affecting bandwidth. See bandwidth versus VARIABLE gain plot below.
The input attenuator connected to the Volts/Div switch is very delicate. Three plastic levers actuate the ×2, ×10, and ×5 attenuator switches. Small metal rollers are clipped into the plastic levers and ride on cams. The plastic clips did not age well and have become very brittle. Combined with lack of lubrication on the metal rollers this often leads to broken plastic clips and missing rollers. Do not operate a stiff Volts/Div switch and lubricate rollers before use!
- Tektronix 7A29 teardown - nice detailed pictures of attenuator internals
- Tek 7A29P @ amplifier.cd including internal pictures
Location of the connectors for the 021-0374-00 GPIB decoder on a 7A29P